View and Download HP A calibration manual online. Synthesizer/Function Synthesizer HP A Operating And Service Manual. Frequency ( pages) . The Operating and Service Manual contains infor¬ mation required to install, operate, test, adjust, and serv¬ ice the Hewlett-Packard Model A. Find great deals for HP a Function Generator Operating and Service Manual . Shop with confidence on eBay!.
|Published (Last):||27 February 2018|
|PDF File Size:||3.59 Mb|
|ePub File Size:||16.96 Mb|
|Price:||Free* [*Free Regsitration Required]|
Service Group K – Output Amplifier Multiply the Ex value by the Ey value and enter in the ExEy space. The following procedures compare the instrument operation to its specifications, listed in Table Cupertino, CA Bourns Inc.
X Drive Linearity Test. The high voltage output is selected by pressing the key in the lower right corner of the front panel. For a negative phase shift, press the key before entering the numerical data. This procedure tests the amplitude of the A ac function output signals against the accuracy specifications in Table For this reason, when operating in Mode 2, the controller can send a program sedvice 48 characters or less to the A, and iYlUUCl vperauun while this data is being processed the controller can unaddress the A to listen and then communicate with another device.
Electronic Counter -hp- Model A, calibrated within three months or with an accurate 10 MHz external reference input a. Operation of any electrical instrument in such an environment constitutes a 3325aa safety hazard.
A to B Frequency Resolution, N. Change A function to triangle. However, if one chooses to modify the sdrvice board to use the older connectors and cablesW36 is required.
Change attenuator to serice dB. Press the Negative Ramp function on the A.
Full text of “HP A Synthesizer/Function Generator Operating and Service Manual”
Two lines will appear on the oscilloscope. The selection of line voltage and fuse is described in Paragraph and Figure When the ac signal is removed in this way, the instrument automatically displays dc offset, and the dc offset entry key light comes on. Amplitude Modulation Envelope Distortion Test. At 18MHz, check for the 3: Failure to comply with these manyal or with specific warnings elsewhere in this manual violates safety standards of design, manufacture, and intended use of the instrument Hewlett-Packard Company assumes no liability for the customer’s failure to comply with these requirements.
Remove the filter and wash thoroughly with eervice water, rinse elcan, and air dry. Service Model A Table The following general instructions 335a be used for repackaging with commercially available materials.
3325A Synthesizer/Function Generator
Integrated Phase Noise Servicd Amplitude is entered and displayed with 4-digit resolution. Verify that ramp period variations do not exceed ten divisions. Sweep Z Blank Output.
NOTE Calculate the values of a j and a 0 to at least five decimal places. If the program string were corrected to make all data valid, this printout would result from the above program.
The A address is normally set at the factory to: English threaded fasteners are colored silver and metric threaded fasteners are colored black.
This removes the ac output. Keyboard and Display, A An amplitude calibration is performed when a register is recalled. Programming Amplitude Units Conversion.
Connect delayed sweep gate from oscilloscope to external trigger input of high speed voltmeter. Frequencies above kHz Model A nn.
Frequency sweep is phase continuous over the full frequency range; that is, there are no discontinuities in the output waveform. Identification of the two types of mounting studs and lockscrews Is made by their color. Connect the A output through a ohm feedthru termination -hp- 1IC to the oscilloscope input input dc coupledor set the A input switch to 50 ohms.
If the value is negative, the first digit is a minus sign. Programming Binary On or Off Functions. See List of Office Locations at the end of this manual. If the instrument fails any of these checks, it should be referred to qualified service personnel for repair. Standard output impedance is 50 ohms.
The instrument related service note is a up directed toward qualified service personnel and is available to all HP Service Centers and customers. Contnillers – Devices that can specify the talker and listener s for an information transfer, The controller can be an active controller or a system controller.
In order to keep the manual up to date, one should periodically request the most recent supplement which is available from the nearest HP Sales and Service Office. Determine the value of a 0 using the equation: Electronic counter should indicate 10 Remove the filter and flush with soapy water, rinse clean, and air dry.
Test Frequency Function Minimum Maximum m Don’t show me this message again.